The writing for the instructions of the Sous Vide Oven controller were finished and the document has 2 720 words and forty pictures. The text of the document was copy/pasted into the Instructables.com’s editor but it was not published until mid-morning to maximize views.
To calibrate the thermometer of the temperature controller some potentiometers were purchased. Multi-turn pots were selected since they can be finely tuned. A 20Ω, a 10KΩ, and a 1MΩ pot were purchased. The 1MΩ pot would only be used if resistance of the thermometer needed to be lowered so it was never used. If this type of calibrating is necessary the potentiometer would be wired in parallel to the thermometer. A short piece of two conductor cable was cut, its ends were stripped, and tinned. The though was that only a very small amount of resistance would need to be added in series. The thermometer wires going to the NTC terminals of the controller were unscrewed. The pot was soldered in series with the thermometer using the middle pot post and the rear post. Al experimentation proved fruitless so the 10KΩ pot replaced the 20Ω pot. The pot was turned to 0Ω and slowly adjusted to come in line with measurements from the IR thermometer. The IR thermometer has .5ºF precision.
I used a cheap macro lens to photograph the 20Ω potentiometer
Wiring to the center post
Wiring to the rear post
The temperature was unable to be lowered using the small pot
The 10KΩ potentiometer replaced the 20Ω pot
The thermometer is cooler than the reading by the temperature controller
Calibrating a $15 controller with a $.25 potentiometer to act like an expensive model
The controller is calibrated
Water was heated and placed into a coffee mug and cool tap water was put into another mug. The heated water was approximately 120°F during testing. The tap water was approximately 79ºF. Some very fine tuning and retesting calibrated to thermometer to within .5ºF which is acceptable since the thermometers used as a reference are not precision instruments. An ideal benchmark for thermometers is 32ºF water with ice cubes but the range of the temperature controller does not go that low nor does it read boiling water.
The warm water tests well
The cool water tests well
Sous Vide Oven
Completed 2014-06-24 (Tu)
Journal page 1
Journal page 2
The rest of the posts for this project have been arranged by date.
A list showing of all the final posts of COMPLETED projects.
Disclaimer for http://24hourengineer.blogspot.com/
This disclaimer must be intact and whole. This disclaimer must be included if a project is distributed.
All information in this blog, or linked by this blog, are not to be taken as advice or solicitation. Anyone attempting to replicate, in whole or in part, is responsible for the outcome and procedure. Any loss of functionality, money, property or similar, is the responsibility of those involved in the replication.
All digital communication regarding the email address firstname.lastname@example.org becomes the intellectual property of Brian McEvoy. Any information contained within these messages may be distributed or retained at the discretion of Brian McEvoy. Any email sent to this address, or any email account owned by Brian McEvoy, cannot be used to claim property or assets.
Comments to the blog may be utilized or erased at the discretion of the owner. No one posting may claim claim property or assets based on their post.
This blog, including pictures and text, is copyright to Brian McEvoy.